On the FARO Laser Scanner, the laser is reflected off the mirror from both the front and back. So, the scanner only rotates 180° along the horizontal axis to capture a 360° scan, as shown in this image. When in Planar View, you may see what looks like a mis-alignment at the point where the scanner starts and stops its rotation. This point is also referred to as the seam.
Here is what you can do to check if a mis-alignment really exists around the seam area.
- Make a selection around the seam area.
- Right-click on the selection and select View > 3D View.
- In the 3D View, rotate around the seam area looking for any mis-alignment.
If the scanner is performing within factory specifications and you see a mis-alignment, it should not exceed the factory-stated tolerances for your scanner. See the FARO SCENE Training Manual and search for Start/Stop Point for more information. You can click the links in the See Also section of this article to view the technical specifications for different Focus3D Laser scanner models.
- Why are there missing scan points near the bottom of the scan area for the Focus3D
- Where can I download the technical specifications for the Focus3D X 30/130/330 and X 130/330 HDR
- Where can I find the technical specifications for the Laser Scanner Focus3D
- Where can I find a FARO Photon Laser Scanner technical specifications
spec, techsheet, tech sheet, accuracy, seam error, degrees, seam gap, collimation error