There is currently a conflict with the Gage 1.5.2 build 198 software and Adobe Flash Player.
- buttons not working during calibration, but working during regular measurements (collecting points)
- Calibration status is listed as "Failed" and never updates
For the software to work properly, you will have to un-install Adobe Flash Player and install the older MacroMedia Flash Player which can be found here. Reboot the computer after installation and run the Gage software again to ensure connectivity.
If you are unable to take points in the probe calibration screen but you are able to take measurements of features, installing the MacroMedia Flash Player should also resolve this issue.
gage, calibration failed, gage software, buttons