- Ensure you are using the correct units (Inches or Millimeters)
- Check for debris under sphere, this will result in the 1" sphere not being centered properly.
- Make sure you are using the calibration method from the training manual, and that the arm is positioned so that its "elbow point" is not sticking straight up.
- Fill out the attached accuracy test form and send to FARO Customer Support.
Click here to download: Acceptance and Accuracy Certification
probe, cal, accuracy